Siemens products may only be used for the applications described in the catalog The SITRANS FUE energy calculator has been developed for Siemens. Fine Controls are the official Technology partner for the Siemens instrumentation product range type FUE, Inline Ultrasonic Energy Meters, In Stock. Dec 4, Siemens Industry Automation Division introduces the Sitrans FUE thermal energy calculator, designed for use in hot water, chilled water.
|Published (Last):||1 June 2009|
|PDF File Size:||5.20 Mb|
|ePub File Size:||5.61 Mb|
|Price:||Free* [*Free Regsitration Required]|
Check cable plug connection of power supply module.
The display menu name is IN0, see menu description Page Service and maintenance 7. The energy calculator is modular in construction and can be fitted with optional modules depending on the application. Reference source not found Page 37 M-Bus or RS module is not used, pulse input module can also be installed in slot 1.
Sitrans Fue And Fus Ufe950 connection of flowmeter, see flowmeter operating instructions.
SITRANS FUE – Automation Technology US – Siemens
A damaged seal will result in immediate invalidation of the factory warranty and calibration. Page 42 Tariff energy 2 1. Page 70 Appendix A. Value of flowmeter pulse output is the same as value of energy meter pulse input.
Page 6 Power supply Only terminals 10 and 11 will be connected to the Siemens flowmeters.
Page 24 pipe by heating low temperature Calculator combined Forward “hot line” By heating: Catalog fke950 instrumentation http: Please talk to your local Siemens contact or follow the return procedures of Siemens Flow Instruments described in chapter: Enter text from picture: Check both type plates for settings.
Don’t show me this message again. Short press 2 sec.
Month loop Display menu loops The sie,ens loops will show the following information: Communication is retained, e. They are intended for trained personnel and therefore do not include basic working steps.
M-Bus or RS module is not used, pulse input module can also be installed in slot 1. The 2 resistors must be mounted between terminals 57 and 58 in the MAG terminal socket. Making cable entries Make a circular hole by dismounting rubber grommet using a flat plier.
Page 56 E – 9 Display Battery empty checked Monitors: For more details, please contact Siemens Flow Instruments. Error is used as a warning. Connect shield to grounding at flowmeter. Type label A typical FUE type label shows the following main information: Remove user sealing if present from integrator housing and open lid. Measurement repeatability stated by manufacturer will be obtained if further described environmental conditions are present.
Page 44 4 sec. RS output Type The optional plug-in RS communication module is a serial interface and permits data exchange with the calculator. Energy calculator is provided with a thread seal or adhesive label seal. Operation hardware Sensors in forward and return lines must fit right to bottom of sensor pockets.
Waste industrial batteries are accepted back by Siemens or by the local Siemens representative. Page 32 The functions and settings of the outputs can be freely programmed using a software tool and the IrDA interface, e. If no mains supply is connected, a back-up battery in the mains unit provides the power supply. Page 21 Flowmeter must be mounted in accordance with the instructions given in the relevant flowmeter operating instructions.
Material compatibility Siemens Flow Instruments can provide assistance with the selection of wetted sensor parts. Ultrasonic flowmeters 7me portable flowmeter pages.
SIEMENS SITRANS FUE – ENERGY CALCULATOR | Fine Controls Ltd
Comments siemena this Manuals Your Name. Scope of delivery may vary depending on selections at ordering. Typically, integrator is mounted on a wall or in a panel. Display LCD off to save power and returns to basic display menu when button is pressed again.
Don’t have an account? In 1 Pulse 1 In 2 Pulse 2 counter counter 6. Figure S – User sealing 3.
Temperature Sensor Dimension drawings