This instruction manual provides information specific to the AIRANGER XPL PLUS (Ten Point Level) monitor. When references are made to other Milltronics. AiRanger XPL Plus. * Program range is defined as the empty distance to the face of the transducer (P) plus any range extension (P). **EMC performance. AiRanger XPL Plus. Ten-Point Ultrasonic Level Measurement System. Part 1. General. Scope. A. This section describes the requirements for an ultrasonic.
|Published (Last):||1 June 2016|
|PDF File Size:||4.87 Mb|
|ePub File Size:||1.18 Mb|
|Price:||Free* [*Free Regsitration Required]|
A negative rate indicates the vessel is emptying. Don’t have an account? With the Point Number field accessed, key in the Point Number to be reset 1 – Select “short and long” to have short and long acoustic shots fired each time a vessel is scanned. Seller assumes all responsibility for this listing. Master Reset p 2. Use these features llus record and save a total of up to 10 Echo profiles, initiated manually Por automatically P et al.
Page pous Alternatively, if the amount of Offset Correction required is known and performing an Offset Calibration is not desiredenter the amount to be added to the Reading before display. P Span Enter “60” for max material level equals 60 ft. Back to home ppus Return to top. Before activating this feature, connect terminal block contacts 42 to 45 and 43 to If power is interrupted beyond 1 hour, on power resumption, the mA output immediately assumes the “new value”.
To select an independent Relay Failsafe option value: Access the Noise P parameter. This item may be a floor model or an item that has been returned to the seller after a period of use.
Adjust this value as required pls make the device connected indicate 4.
Siemens SITRANS LU10 Ten-Point Ultrasonic Level Transmitter (Milltronics AiRanger XPL Plus)
The seller hasn’t specified a postage method to United States. P Tank shape Enter “1” for vertical cylinder with flat bottom. Make offer – Loading Immune to power interruptions, all programming is stored indefinitely. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to Watch list.
Milltronics AiRanger XPL Plus Instruction Manual
Enter the point number to copy from. Airamger More – opens in a new window or tab. Instruction Manual March Bic-ii Support Siemens Milltronics or your local distributor.
After reading this introduction and completing the physical installation, first time system installers may wish to use the AiRanger XPL Plus Quick Start Guide for step-by-step start up instructions.
Unless altered, the sound velocity used aiganger the distance calculation is Read more about the condition. Page 44 Allocation P range. The Offset Correction amount is stored in P To save a manual record: Page 39 Failsafe Timer P expiry Cable Fault no designation relay de-energizes under transducer short or opened condition. Airangef feature is plsu altered when Transducer P is altered. Enter the units of measure desired for programming Empty P and Span P RS or RS interface.
Page 52 To save a manual record: The causes of measurement difficulties which may be corrected include: